The flow meter will produce by order based on the condition used.

In the case of an estimate, please direct the measured fluid name, range of flowing quantity, pressure, temperature,etc.

Technological data | precision | withstand pressure/ leakage inspection | flow correction formula |

### It is necessary for measuring liquid (specific gravity and viscosity).

When the liquid of the water corresponding is measured, it is of no matter, but we be designed and product with specific gravity and viscosity. A temperature variation is a big influence is exerted on the display of flowing quantity, so please keep a fixed temperature as much as possible.

### It is necessary for measuring gas (pressure and temperature).

we will set the condition of the measured gas (pressure and temperature) and produce

the flow meter for the gas.

The flow quantity becomes flow quantity converted into normal condition (atmospheric pressure 20 degrees Celsius).

In addition, we produce even a standard state (atmospheric pressure 0 degrees

Celsius) by a demand.

Fig. A, pressure 0.3MPa joins the needle valve and figure A becomes a state (atmospheric pressure) without pressure drag after it.Setting pressure is atmospheric pressure (1atm).
Fig. B, Pressure 0.3MPa joins the flow meter and the needle valve, and even when be that the atmospheric pressure or some load pressures joins after that, set pressure becomes 0.3MPa. Please always do setting pressure in the state that constant pressure hung over to a flowmeter by pressure when gas to measure passes by in a flowmeter.In addition, when pressure and the temperature change, it will correct it by the value of fluctuations. |

### Precision

The principle of accuracy is within based on JIS ±2%- ±5% of maximum scale value (F.S). However, accuracy might decrease when it is the following.

- Minute flow quantity
- Liquid of high viscosity
- Gas with extremely small density

### Withstand pressure/ leakage inspection

It does by the pressure 1.5 times the maximum use pressure.

### Flow correction formula

**Liquid correction formula**

When the density is different in the different kind liquid | |

### Different kind gas, Pressure and temperature correction formula

When the unit of pressure is kPa | |

When the unit of pressure is kgf/cm2(G) | |

### Steam correction formula

*In general, the steam shows Mass(weight) by the unit |

### Explanation of sign

Q |
： | True flow quantity |

q |
： | Flow quantity on a dial |

ρｆ |
： | Density of a float |

ρ0 |
： | Density of a fluid on a dial |

ρ1 |
： | Density of a different fluid |

Ｐ0 |
： | Pressure (kPa/kgf/cm2G) on a dial |

Ｐ1 |
： | Different pressure (kPa/kgf/cm2G) |

Ｔ0 |
： | Temperature on a dial |

Ｔ1 |
： | Different temperature |

Ｗ1 |
： | True flow quantity (mass) |

Ｗ0 |
： | Flow quantity (mass) on a dial |

γ0 |
： | Density of steam on a dial |

γ1 |
： | Density of different steam |